The ball type load cell is easy to install and does not require on-site calibration/corner correction
Digital O/P sends weight directly in steps of 1 kg in digital format, wherein the analog load cell, the output is m/v without analog signal
Since the output is a digital output, the problems related to the load cell cable, such as moisture and temperature, are completely eliminated. This is the most important advantage compared to a floor scale using an analog load cell
In the load cell manufacturer's digital load cell, the different lengths of the load cell cable of different lengths will not affect the calibration and turning, but this is very critical in the analog load cell. Very high stability. The greater the number of load cells in the weighing bridge, the better its resolution and accuracy, because the number of each load cell is added together. In the analog load cell, the output of each load cell is averaged (integrated) in the junction box, and then the effective output is input to the analog indicator.
Simple after-sales service: If one of the analog load cells in the weighbridge fails, the complete wiring must be removed from the conduit and junction box to find the fault. In the digital mode, the output of each load cell is displayed on the indicator separately, so it is easy to find faults.
Plug-and-play adaptability: In the weighbridge based on analog load cells, if a single load cell is replaced, turning and calibration are required again. Therefore, this is a very expensive and time-consuming operation because it will require placing two-thirds of the minimum weight on the platform, and in a digital load cell, it can be replaced without affecting turning and calibration.
The above points clearly show that any technician who uses digital load cells can replace the load cell, while analog load cells cannot, so there is no need for expensive and experienced service engineers, and it is also equipped with diagnostic LEDs